GE Healthcare c Anesthesia Machine Technical Reference Manual Software Revision 1.x Approved −12−9 EET Junmin Ji (Rick) M c Product Data Sheet – GE Healthcare c Familiar, reliable, GE – with over years experience in anesthesia and quality leadership? 53 cm/21 in Depth: 40 cm/16 in Side of machine: cm/45 in Height: The c is the ideal solution for customers seeking an affordable, reliable and easy-to-use anaesthesia system. The c is designed and manufactured in.
|Genre:||Health and Food|
|Published (Last):||13 July 2010|
|PDF File Size:||19.81 Mb|
|ePub File Size:||1.35 Mb|
|Price:||Free* [*Free Regsitration Required]|
Paw Gain Section 4. The flow may be temporarily increased to facilitate the decay. Make sure to push the new check valve all the way back into the opening until it bottoms out on the shoulder. Perform the following steps: N2O pipeline inlet Pneumatic Vent Engine 1. Lift the upper shelf. Remove the rear upper panel Section 9.
To replace the backlight annesthesia. Remove the caster with an appropriately sized open- end wrench. Disconnect the leads from the old switch and reconnect them to the new switch. Bleed all gas pressure from the machine Section 9. Trim the tail flush with outside surface of the valve seat refer to kachine removed flapper. Lubricate the o-ring sparingly with Krytox. Disconnect the wires from the ACGO mode microswitch 4 at the back of the selector switch. If necessary, access the Service Mode — Languages menu and select the correct language.
While holding the switch assembly, push in the knob and turn it counterclockwise. GE Healthcare 9100c Familiar, reliable, affordable Trust points?
c NXT | GE Healthcare
To remove the lower access panel, fully loosen the six screws that hold the panel in place. Check the switch actuator lever to ensure free movement. Test the replacement switch assembly: Help Center Find new research papers in: Breathing system tests Section 3. If leak persists, replace the flapper. Lubricate the o-ring sparingly with Krytox do not get Krytox on the flapper.
Remove the pressure test device, test adapter, patient circuit and mzchine manual bag.
Multipurpose Anaesthesia Machine- C – Yoga Enterprises, Pune | ID:
O-ring Nipple Seal Spring O-ring 6. Mcahine the pipeline supplies and close all cylinder valves if equipped. Remarks The data is stored in EEPROM and is used during normal operation to compensate for the individual flow sensor output characteristics.
Failure to do so can anestbesia in damage to the flow valve and possible injury to the patient. Test the high airway pressure alarm: The data is stored in EEPROM and is used during the calibration to compensate for the individual calibration orifice characteristics. Install the module cover.
Hypoxic guard system Type: Active AGSS systems with a Combines clinical performance with ergonomic design to deliver anestheic agents effectively and efficiently.
Do not over tighten the screw securing the reservoir.
Remove the Software USB disk. The flush valve supplies high flows of O2 to the common gas outlet when you push the flush button. The following illustration shows the parts. Use the following tightening procedure whenever you are replacing a cylinder supply or pressure tube fittings a cylinder wnesthesia gauge.
Click here to sign up. Make sure a low O2 alarm occurs. Turn all flow controls fully clockwise closed. Remove the upper rear panel Section 9. This Check tubing connections. Therefore, GE agrees to make available Interfacing 900c and data, as described in the Interfacing Commitment, between the following product combinations:. Nature of Macnine Manufacturer. Pressure gauge Scale range: Reassemble the bellows assembly upon the completion of calibration.
Low Pressure Leak Vaporizer not installed properly Reseat vaporizer if not installed properly.