The Corus PTZ is a key element in the whole Actaris chain, from the meter to the billing data, because it converts the actual volume measured. For each FLUXI TZ and DELTA meters, correction according to pressure, temperature and compressibility factor (meaning the transformation in real. CORUS is an electronic volume converter dedicated to commercial and industrial applications. It converts the actual volume measured by the gas meter to.
|Published (Last):||1 May 2006|
|PDF File Size:||15.14 Mb|
|ePub File Size:||12.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
Select the type of alarm to be resetted: Period 1 second Retrans. Program the required parameters Keyboard or communication?
Itron Corus Gas Volume Converter
Its four wires mounting provides a high accuracy by pyz the resistance of the cable. Date and time of the event? Status The event log has a capacity of events and cannot be reset.
According to the European regulation, it bears the CE marking actarsi complies with the following directives: These two actions are mandatory to achieve a correct water tightness of the enclosure. We would like to thank all our customers for your continued support over the last 5 years.
Pb T Z The reference pressure Pb and the reference temperature Tb must be programmed into the device. For each event, following data are available: The system is built in an IP65 enclosure, for wall or meter mounting. These plates are screwed and sealed.
Enable our cloud-based energy analytics and dashboard for your gas meter readings, to discover, measure and verify savings in the premises you manage. Two 2 operating pressure range are available: The corresponding marking is: Both pulse outputs are individually characterized by a pulse weight that can be configured to any value greater than or equal to the input pulse weight among 0. The various LF cable Binder plug available are: If the function is activated, an alarm of the above list is set if the corresponding flow-rate gets higher or lower than the threshold.
When the connected device initiates a communication, it has to take in account a wake up time of a few milliseconds. To prevent any damage, following operations are recommended to limit electrostatic discharges on the enclosure of the CORUS: This alarm is not metrological and is dedicated to the control of the whole metering chain.
The frequency of retransmissions indicated in the table actarks to average frequency on the 2 outputs, when both channels are configured ltz pulses.
E-Gaz Malzemeleri Mühendislik Müşavirlik Sanayi ve Ticaret Ltd. Şti.
The outputs generate a pulse every time the corresponding unconverted or converted index gets increased by a value equal to the selected output pulse weight.
For safety reasons, it is strictly forbidden to use other modules than those specified by ‘Actaris’.
All daily consumptions are stamped from the gas hour to the gas hour of the next day. The Octopus gateway is a Modbus data logger that comes with 6 x digital inputs, meaning that up to 6 pulses outputs can be counted, logged and communicated to Wattics dashboard. Base volume outputs are automatically disabled as long as any metrological alarm is active. The actafis input can be put into one of corks following modes: This data is protected by the metrological switch.
It is delivered with a cable length of either 2.
Solenvis SL79 Itron-Corus gas volume converter
It must be preferentially screwed to the “Pr” reference pressure tap of the meter: Our team will get back to you shortly. Qualified personnel are those who, based on their training and experience, are capable of identifying risks and avoiding potential hazards when working with this product.
The temperature measurement period is programmable from 1 to 20 seconds. Bank of England select Solenvis to vorus clamp-on metering.
Pbase, Tbase, Gas composition according formula: After 2 minutes with no action on the keyboard, the display switches off. The pressure sensor intends two 2 holes allowing to seal and link the sensor with the meter installation.
You may have a signal barrier in between to separate the safe zone from the hazardous zone. The converter must then be configured into the ‘external power supply’ mode. Input pulse weight, unconverted Index, Converted Index: This port can be used for serial asynchronous start-stop half-duplex communications. There displayed sequence is based on five main menus: